KV4.3/KChIP2 - Dose-response curve of Flecanaide
Screenshots of the PatchControl 384 software showing KV4.3/KChIP2 current traces in response to a voltage step protocol and the corresponding current-voltage relationship plot. Using whole cell mode in combination with multi-hole chips (4 holes per well), stably transfected cells were measured on the SyncroPatch 384PE (a predecessor model of SyncroPatch 384). The IC50 value of flecainide was determined as 28.3 µM which is in accordance to literature. The success rate for completed experiments was 100%.